- Research article
- Open Access
Kinect-based choice reaching and stepping reaction time tests for clinical and in-home assessment of fall risk in older people: a prospective study
European Review of Aging and Physical Activity volume 13, Article number: 2 (2016)
Quick protective reactions such as reaching or stepping are important to avoid a fall or minimize injuries. We developed Kinect-based choice reaching and stepping reaction time tests (Kinect-based CRTs) and evaluated their ability to differentiate between older fallers and non-fallers and the feasibility of administering them at home.
A total of 94 community-dwelling older people were assessed on the Kinect-based CRTs in the laboratory and were followed-up for falls for 6 months. Additionally, a subgroup (n = 20) conducted the Kinect-based CRTs at home. Signal processing algorithms were developed to extract features for reaction, movement and the total time from the Kinect skeleton data.
Nineteen participants (20.2 %) reported a fall in the 6 months following the assessment. The reaction time (fallers: 797 ± 136 ms, non-fallers: 714 ± 89 ms), movement time (fallers: 392 ± 50 ms, non-fallers: 358 ± 51 ms) and total time (fallers: 1189 ± 170 ms, non-fallers: 1072 ± 109 ms) of the reaching reaction time test differentiated well between the fallers and non-fallers. The stepping reaction time test did not significantly discriminate between the two groups in the prospective study. The correlations between the laboratory and in-home assessments were 0.689 for the reaching reaction time and 0.860 for stepping reaction time.
The study findings indicate that the Kinect-based CRT tests are feasible to administer in clinical and in-home settings, and thus represents an important step towards the development of sensor-based fall risk self-assessments. With further validation, the assessments may prove useful as a fall risk screen and home-based assessment measures for monitoring changes over time and effects of fall prevention interventions.
Falls are an important problem in older people and a major public health issue. A range of physiological impairments including poor balance [1, 2], impaired gait [3, 4], muscle weakness [5, 6] and slow voluntary reaction times [7–9] have been associated with falls.
Quick protective reactions that involve reaching or stepping movements are important to avoid falls  or to reduce the risk of severe injuries [11, 12]. Tests that reveal deficits in upper- or lower-limb responses may help identify older people at risk of falls or increased risk of fall injury because of poor protective responses after a loss of balance has occurred [10, 13].
Previously, simple and choice reaction times with finger- or foot-press responses have been measured with electronic timers and switches in the laboratory [7, 9]. Long term monitoring by regularly repeated assessments would allow identifying the risk of falling over time or the tracking of improvements following an exercise program. But, because of limited resources of most healthcare systems, regularly repeated assessments are not feasible. Inexpensive, easy to administer, portable and accurate reaction time tests would make it possible to incorporate these tests in clinical practice or even self-assessments performed at home.
Recent advances in sensor technologies and research in human computer interaction hold great promise for these new methods of fall risk assessment [14, 15]. However, to date, only a few studies have focussed on the development of a home-based reaction time test for fall risk assessment. In one study a choice reaction time test using an infrared laser in combination with a plus-shaped mat was developed to measure stepping responses to optical cues . Another stepping study evaluated a mat-based system with pressure sensors to assess and improve stepping responses [17, 18]. To our knowledge, no research exists on home-based upper-limb reaction time tests.
The aim of this study was to examine the potential of Kinect-based, portable and low-cost assessment tests of choice reaching and stepping reaction time (referred as Kinect-based CRTs) to assess fall risk. The main aims of the study were to: (i) investigate whether the Kinect-based CRTs could differentiate between older fallers and non-fallers and (ii) examine the feasibility of conducting the CRT tests in a home setting.
A total of 94 community-dwelling older people living in retirement villages in Sydney, Australia participated in this study. The sample was drawn from two trials: 1) iStoppFalls trial (ACTRN12614000096651)  and 2) SureStep trial (ACTRN12613000671763). The inclusion criteria were: living independently (i.e. not in assisted living or nursing homes), aged 65 years or older and being ambulant with or without the use of a walking aid. The exclusion criteria were: medically unstable, suffering from major cognitive impairment (Mini-Cog < 3), neurodegenerative disease or colour blindness. Written informed consent was obtained from all participants prior to data collection. The study was approved by the University of New South Wales Human Studies Ethics Committee.
Kinect-based choice reaction time tests (Kinect-based CRTs)
The Kinect-based CRTs are tests which rely on video-based motion capture technology (i.e. Microsoft Kinect). They comprise 1) a choice reaching reaction time test (Fig. 1a) and 2) a choice stepping reaction time test (Fig. 1b). When conducting the Kinect-based CRT tests participants see themselves represented as an avatar in a virtual environment on a TV screen. The tests start with the participant standing in a normal comfortable position with the arms by the side. Two lights, one to the left and one to the right side of the avatar, flash up in random order. In the reaching reaction time test, participants are instructed to lift their corresponding arm to the flashing light as fast as possible. In the stepping reaction time test, participants have to take a step onto the flashing light, using the left foot when the left light flashes and the right foot when the right light flashes, as quickly as possible.
The study protocol included the following parts:
Laboratory assessment: All participants were assessed on the Kinect-based CRTs and on clinical tests for reaction time and fall risk.
Association with future falls: Participants were followed up for falls for 6 months and the ability of the Kinect-based CRT tests to differentiate between the fallers and non-fallers was investigated.
In-home assessment: Following the laboratory assessment the Kinect-based CRTs were conducted with a subgroup of participants at home and the relationships between the laboratory and in-home assessments were analysed.
All participants were initially assessed with the Kinect-based CRT tests in the laboratory. For each participant 40 reaching and stepping responses were recorded with a short break of less than a minute after 20 responses. The first five trials were practice trials and excluded from data analysis. The assessments were video recorded with two video cameras (i.e. front and side view) to support the researchers during the data analysis process. The Physiological Profile Assessment (PPA) was conducted as an estimate of the overall fall risk of the participants. The PPA is based on tests which assess sensorimotor abilities: balance (sway when standing on medium-density foam with eyes open), lower extremity muscle strength (knee extension), contrast sensitivity (Melbourne edge test (MET), peripheral sensation (proprioception) and single hand (i.e. finger-press) reaction time .
The convergent validity of the Kinect-based CRT tests in relation to the simple reaction time of the PPA and choice reaction time of the Attention Network Test (ANT) were examined. The ANT is a computer-based test where participants had to determine whether a central arrow points to the left or right and to press the corresponding button on a PC-keyboard as quickly as possible .
Association with future falls
Participants were followed-up for 6 months and asked to report their falls with monthly falls calendars. Follow-up telephone interviews were conducted if participants failed to return their calendars. A fall was defined as ‘an unexpected event in which the person comes to rest on the ground, floor, or lower level’ . Participants were classified as fallers if they experienced at least one fall in the 6 months follow-up period.
The feasibility to administer the Kinect-based CRTs at home was examined in a subsample of 20 participants. The system was installed in the participants’ homes and the CRT tests were conducted under supervision of a trained researcher. The time gap between the laboratory assessment and the in-home assessment was on average 40 (±20) days.
Data acquisition and analysis
The Microsoft Kinect is a marker-free computer vision sensor that can measure three-dimensional motion of a person. In the laboratory, the Kinect sensor was placed in front of the TV screen at a height of 80 cm and a distance of 200 cm from the participants. Skeleton data of anatomical landmarks in world coordinates were recorded using the Kinect Software Development Kit for Windows with a sampling rate of 30 Hz and a resolution of 640 × 480 pixels.
For the Kinect-based CRTs the horizontal displacement data (i.e. movements in the x-axis to the left or the right) of the Microsoft Kinect sensor were used for the algorithms. In detail, the skeleton data of the left and right hand tracking were obtained for the reaching reaction time test and the tracking data of the feet for the stepping reaction time test. The signals were low-pass filtered using a 4th order Butterworth filter with a cut-off frequency of 2 Hz to reduce noise. The following features were automatically extracted from each recording (Fig. 2):
Reaction time: The reaction time was defined as the time from the cue signal until the first movement of the hand or foot. The movement initiation was detected as a change in position of at least 5 cm (i.e. to the left or right) compared to the rest position. The mean across all reaction times was calculated.
Movement time: The movement time was defined as the time from the movement initiation until the corresponding virtual target was hit by the hand or foot. Incorrect movements for example in the opposite direction of the cue signal were excluded. The mean movement time across all correctly identified movements was calculated.
Total time: The total time was defined as the sum of the reaction and movement time.
One-way ANOVA was used to evaluate mean differences in the test measures between the fallers and non-fallers. Pearson’s correlation coefficients were calculated to quantify convergent validity and the relationship between the laboratory and in-home assessments. Correlation results were categorized as weak (0.1 – 0.3), moderate (0.4 – 0.6), and strong (0.7 – 0.9) . P-values less of 0.05 were considered to be statistically significant. Signal processing, data analysis and statistical analysis were performed in MATLAB 8.2 (R2013b).
Ninety-four persons (62 women) aged 80.6 ± 6.9 years participated in the study. On average, participants were 163.7 ± 9.8 cm tall, weighed 72 ± 15.2 kg, had a Body-Mass-Index (BMI) of 26.8 ± 4.7 and PPA fall risk score of 1.48 ± 0.88 indicating a moderate risk of falls .
The Kinect-based reaching reaction time was significantly correlated with the simple reaction time of the PPA (r = 0.338, p < 0.001) and choice reaction time of the ANT (r = 0.593, p < 0.001). Similarly, the Kinect-based stepping reaction time measurement was significantly correlated to the PPA reaction time (r = 0.403, p < 0.001) and ANT reaction time (r = 0.576, p < 0.001) tests.
Association with future falls
Nineteen participants (20.2 %) reported one or more falls in the 6 months following the assessment. There was no significant difference in age, height, weight or BMI between the fallers and non-fallers. Fallers were significantly slower than non-fallers on the reaching reaction time test measurements (Table 1). The stepping reaction time test, the simple reaction time of the PPA assessment and ANT choice reaction time did not significantly discriminate between the groups.
The in-home assessments were conducted with 20 participants (14 women, 2 fallers). Figure 3 illustrates the linear relationships between the laboratory and in-home assessments. On average, the Kinect-based reaching reaction time was 772 ms ± 85 ms at home compared with 771 ms ± 139 ms in the laboratory (p = 0.987). The correlations were moderate to strong for the reaction time (r = 0.689, p < 0.001), movement time (r = 0.505, p = 0.023) and total time (r = 0.737, p < 0.001). Similarly, there was no significant difference between the Kinect-based stepping reaction time at home with 862 ms ± 144 ms and in the laboratory with 876 ms ± 215 ms (p = 0.594). The correlations were strong for the reaction time (r = 0.860, p < 0.001) and total time (r = 0.814, p < 0.001). Movement time was moderately correlated (r = 0.609. p = 0.004).
This study, examined the feasibility of Kinect-based reaching and stepping reaction time tests. Signal processing algorithms were developed to quantify performance on these tests and the convergent and discriminant validity of the derived sensor-based measurements were evaluated. To our knowledge, this is the first study using a Kinect-based approach to assess upper- and lower-limb reaction time conducted in both the laboratory and home in community-dwelling older people.
We found that fallers were slower than non-fallers on the reaching reaction time test measurements. Our finding is consistent with the results of previous studies showing that slow reactions are associated with an increased risk of falling [10, 23]. Use of the upper-limbs is a common response to prevent a fall and to reduce risk of injuries [10, 11]. It has recently been reported that in frail older people the protective responses are often ineffective because of lack of strength and movement speed . The Kinect-based CRT may help to reveal deficits in upper-limb responses for targeted improvement.
In a previous study, a test of stepping reaction time was shown to discriminate well between older fallers and non-fallers based on 12 month fall history , but this could not be verified in this prospective study, possibly because of the short follow-up period. We believe the Kinect-based stepping reaction time test has some benefits when compared to other approaches. Other systems require a step mat placed on the floor. This could be seen as an advantage as it provides physical targets during stepping. However, step mats also pose a potential trip hazard, require more time to set up and limit movements to the pre-defined fields.
The reaction times measured with the Kinect-based CRTs were on average longer compared to measurements from traditional tests using electronic timers and switches [7, 9]. However, when compared the results to studies using similar technical equipment the reaction times were almost identical [16, 17]. This can be explained with a delay of the TV and Kinect sensor for data acquisition, video processing and display. Future studies are warranted to investigate the relevance of this possible measurement error.
The advantages of a Kinect-based system are 1) easy to set up - no further physical equipment is needed, 2) safe - no additional trip hazards, 3) inexpensive - the Microsoft Kinect is a widely available consumer device and 4) fairly accurate [25, 26] - enables whole body tracking of participants’ movements. These characteristics enable its use in a clinical setting or even in the homes of the older people as an assessment or training tool. Currently, regular repeated assessments are not feasible in clinical practice and therefore the assessments are often only weakly associated with falls. The correlations between the Kinect-based CRT laboratory and in-home assessments were moderate to strong which suggests the conduct of the Kinect-based CRT tests at home is feasible. Noteworthy, the correlations were stronger for the reaction times compared to the movement times. This could indicate that the reaction time measurements are less dependent on the environmental conditions (e.g. distance to the Kinect sensor, size of TV).
We acknowledge certain study limitations. The in-home assessment was conducted with a relatively small subgroup of participants and that might limits the generalizability of the results. Furthermore, for organisational reasons, all 94 participants had to be tested in the laboratory first, before the Kinect-based CRTs could be conducted in the private homes. This resulted in a longer time gap between the laboratory and home assessments for people who were assessed early compared to people who were assessed later in the study. The sample size was only moderate for a fall risk study and the prospective follow-up period relatively short for capturing sufficient fall events. Larger-scale studies are therefore necessary to confirm the presented results. However, our encouraging findings suggest the Kinect-based CRTs could be conducted by the following means 1) tests performed in a clinical setting, 2) tests administered in regular home visits by trained personnel or 3) tests performed independently and unsupervised as self-assessments.
In summary, our findings indicate the Kinect-based CRT tests are feasible to administer in clinical and in-home settings, and thus represents an important step towards the development of sensor-based fall risk self-assessments. With further validation, the assessments may prove useful as a fall risk screen and home-based assessment measures of upper- and lower-limb movements for monitoring changes over time as well as the effects of fall prevention interventions.
Lord SR, Rogers MW, Howland A, Fitzpatrick R. Lateral stability, sensorimotor function and falls in older people. J Am Geriatr Soc. 1999;47:1077–81.
Overstall PW, Exton-Smith AN, Imms FJ, Johnson AL. Falls in the elderly related to postural imbalance. Br Med J. 1977;1:261–4.
Maki BE. Gait changes in older adults: predictors of falls or indicators of fear. J Am Geriatr Soc. 1997;45:313–20.
Hausdorff JM, Rios DA, Edelberg HK. Gait variability and fall risk in community-living older adults: A 1-year prospective study. Arch Phys Med Rehabil. 2001;82:1050–6.
Moreland JD, Richardson JA, Goldsmith CH, Clase CM. Muscle weakness and falls in older adults: a systematic review and meta-analysis. J Am Geriatr Soc. 2004;52:1121–9.
Campbell AJ, Borrie MJ, Spears GF. Risk factors for falls in a community-based prospective study of people 70 years and older. J Gerontol. 1989;44:112–7.
Lord SR, Fitzpatrick RC. Choice stepping reaction time: a composite measure of falls risk in older people. J Gerontol. 2001;56:627–32.
Pijnappels M, Delbaere K, Sturnieks DL, Lord SR. The association between choice stepping reaction time and falls in older adults--a path analysis model. Age Ageing. 2010;39:99–104.
Lord SR, Menz HB, Tiedemann A. A physiological profile approach to falls risk assessment and prevention. Phys Ther. 2003;83:237–52.
Maki BE, McIlroy WE. Control of rapid limb movements for balance recovery: age-related changes and implications for fall prevention. Age Ageing. 2006;35:12–8.
Nevitt MC, Cummings SR, Hudes ES. Risk factors for injurious falls: a prospective study. J Gerontol. 1991;46:164–70.
Feldman F, Robinovitch SN. Reducing hip fracture risk during sideways falls: Evidence in young adults of the protective effects of impact to the hands and stepping. J Biomech. 2007;40:2612–8.
Schonnop R, Yang Y, Feldman F, Robinson E, Loughin M, Robinovitch SN. Prevalence of and factors associated with head impact during falls in older adults in long-term care. Can Med Assoc J. 2013;185:803–10.
Ejupi A, Lord SR, Delbaere K. New methods for fall risk prediction. Curr Opin Clin Nutr Metab Care. 2014;17:407–11.
Howcroft J, Kofman J, Lemaire ED. Review of fall risk assessment in geriatric populations using inertial sensors. J Neuroeng Rehabil. 2013;10:1–12.
Nishiguchi S, Yamada M, Uemura K, Matsumura T, Takahashi M, Moriguchi T, et al. A novel infrared laser device that measures multilateral parameters of stepping performance for assessment of fall risk in elderly individuals. Aging Clin Exp Res. 2013;25:311–6.
Schoene D, Smith ST, Davies TA, Delbaere K, Lord SR. A Stroop Stepping Test (SST) using low-cost computer game technology discriminates between older fallers and non-fallers. Age Ageing. 2013;43:285–9.
Schoene D, Lord SR, Delbaere K, Severino C, Davies TA, Smith ST. A randomized controlled pilot study of home-based step training in older people using videogame technology. PLoS One. 2013;8:1–8.
Gschwind YJ, Eichberg S, Marston HR, Ejupi A, De Rosario H, Kroll M, et al. ICT-based system to predict and prevent falls (iStoppFalls): study protocol for an international multicenter randomized controlled trial. BMC Geriatr. 2014; 14:1–13
Fan J, McCandliss B, Sommer T, Raz A, Posner MI. Testing the efficiency and independence of attentional networks. J Cogn Neurosci. 2002;14:340–7.
Lamb SE, Jørstad-Stein EC, Hauer K, Becker C. Development of a common outcome data set for fall injury prevention trials: the Prevention of Falls Network Europe consensus. J Am Geriatr Soc. 2005;53:1618–22.
Dancy CP, Reidy J. Statistics without maths for psychology. Harlow: Pearson Education Limited; 2004.
Lajoie Y, Gallagher SP. Predicting falls within the elderly community: Comparison of postural sway, reaction time, the Berg balance scale and the Activities-specific Balance Confidence (ABC) scale for comparing fallers and non-fallers. Arch Gerontol Geriatr. 2004;38:11–26.
Ejupi A, Brodie M, Gschwind YJ, Schoene D, Lord SR, Delbaere K. Choice Stepping Reaction Time test using Exergame technology for fall risk assessment in older people. In Proceedings of the IEEE Conference on Engineering in Medicine and Biology: September 2014; Engineering in Medicine and Biology Society (EMBC): Chicago.
Clark R, Pua Y-H, Fortin K, Ritchie C, Webster KE, Denehy L, et al. Validity of the Microsoft Kinect for assessment of postural control. Gait & Posture. 2012;36:372–7.
Galna B, Barry G, Jackson D, Mhiripiri D, Olivier P, Rochester L. Accuracy of the Microsoft Kinect sensor for measuring movement in people with Parkinson’s disease. Gait & Posture. 2014;39:1062–8.
The PPA (NeuRA FallScreen) is commercially available through Neuroscience Research Australia. The authors declare that they have no further competing interests.
AE collected the data, developed the signal processing algorithms, analyzed the data and wrote the manuscript. YG helped with the data collection. All authors were involved in the conception and design of the study. All authors discussed the results and implications and commented on the manuscript at all stages. All authors read and approved the final manuscript.
About this article
Cite this article
Ejupi, A., Gschwind, Y.J., Brodie, M. et al. Kinect-based choice reaching and stepping reaction time tests for clinical and in-home assessment of fall risk in older people: a prospective study. Eur Rev Aging Phys Act 13, 2 (2016). https://doi.org/10.1186/s11556-016-0162-2
- Older adults
- Accidental falls
- Fall risk assessments
- Microsoft kinect
- Reaction time